The APOGEE Northern and Southern Spectrographs
The first APOGEE spectrograph was constructed and assembled primarily at the University of Virginia. It is situated at the 2.5-m Sloan Foundation Telescope at Apache Point Observatory and was commissioned in August 2011 during the SDSS-III APOGEE-1 Survey period. The second APOGEE spectrograph had a similar build process and is almost analogous to the first. It is located at the 2.5-m Irenee du Pont Telescope at Las Campanas Observatory and was commissioned two and half years into the SDSS-IV APOGEE-2 Survey, during February and March of 2017.
For the set-up of both the Northern and Southern instruments, light is transmitted telescope focal plane to the pseudo-slit within the cryogenically cooled instrument via 300 low-OH ("dry") fused silica fibers which have a 2"(N)/1.3"(S) field of view on the sky. Each of the 300 fiber runs consist of two fiber assemblies connected in series. A 2-m fiber run (so-called "fiber harness") goes from the plug plate to a "gang-connector" just below the telescope and plug plate cartridge. Multiple cartridges are used throughout the night to observe different parts of the sky and each cartridge has its own set of 300 fiber harnesses. An innovative gang-connector allows the simultaneous connection of all 300 fibers from a specific cartridge to the single fiber run ("fiber link") which transmits the light from the telescope and cartridges into the APOGEE spectrograph. To avoid throughput losses from the use of another fiber coupling, the fibers are fed through an epoxy-sealed vacuum feed-through without break at the cryostat wall. Lastly, fibers that make up the fiber link terminate at the instrument pseudo-slit inside the cryogenically cooled instrument.
An 'uncorrected' Schmidt camera, used in reverse, collimates the light of each of the fibers. The fiber tips are carefully positioned on a curved pseudo-slit. The pseudo-slit and spherical collimator have a common center of curvature near the system pupil which is also the approximate position of the spectrograph grating. The design is on-axis so the pseudo-slit is an obscuration in the collimated beam. Two-fold mirrors are used for efficient packaging of the optics train within the cryostat.
The dispersive optic is a transmissive 3-panel mosaic Volume Phase Holographic (VPH) grating fabricated by Kaiser Optical Systems Inc., the first ever of its kind deployed in a cryogenic astronomical instrument (note for the Southern instrument, there is a slight departure in that the VPH is 2-panel). Due to its size, the grating area of the VPH was recorded in multiple steps (panels) and then processed and capped as a single unit. A 6-element refractive camera fabricated by New England Optical Systems focuses the various wavelengths of light onto the detectors. The camera is composed of mono-crystalline silicon and fused silica elements, the largest of which are nearly 400 mm diameter. Three JWST H2RG near-infrared detectors, on-loan from the University of Arizona, are mounted side-by-side to record the blue, middle and red portions of the spectrum. An Astronomical Research Camera (so-called Leach) controller operates all three detectors in sample-up-the-ramp mode.
While the nominal full-width half-maximum is approx. 2.3 pixels wide, the blue end of the spectrum is sampled with less than 2 pixels. To recover optimal sampling, the detector mount is translated (spectrally dithered) between sets of frames with a custom, precision single-axis actuator. For further details regarding the APOGEE spectrographs, users should consult the forthcoming publication from Wilson et al. (in prep.).
General Characteristics of the Northern and Southern Spectrographs
- Spectral resolution
- 22,500 (approx.)
- Wavelength coverage
- 1.51 - 1.70 μm
- Fiber diameter
- 2 arcsec (Northern Spectrograph), 1.31 arcsec (Southern Spectrograph)
- ~15% H broad-band efficiency (including atmosphere)
- S/N~100/pixel for H < 12.2 and 3-hour integration
- Three JWST H2RG (2048 x 2048) Near-Infrared HgCdTe Detectors with 18 micron pixels